Dry thawing device for the thaw of Fresh Frozen Plasma (FFP), Cryoprecipitated AHF, Whole Blood, Red Blood Cells (RBC) and crystalloid infusion solutions.
Also written into the Novartis patent for Kymriah.
To make optimum use of the Barkey plasmatherm, up to four blood or plasma bags are placed between two cushions filled with water. During this process, the water is pumped into the cushions where it circulates continuously. This dry tempering process guarantees constant temperatures with minimal variations in accordance with the applicable guidelines for blood and blood products,,,.
Product safety in accordance with hygiene guidelines is guaranteed, as the bag preserves are not in direct contact with the heating medium water (dry tempering). The cushions ideally conform to the shape of the preserves. Integrated moisture sensors detect leaks at an early stage and use visual and audible signals to alert the user to possible deviations. Visual inspection of the bag preserves by means of a transparent lid unit enables the monitoring of sensitive blood and blood products. The safety of the product and its biological value are permanently guaranteed by a preset temperature limitation.
The user can choose between 4 preset programs for Blood, Plasma (FFP), Continuous and a special User program.
The Barkey plasmatherm also stands for sustainability. The product does not require any additional consumables. In accordance with the latest quality standards, the Barkey plasmatherm is manufactured according to ISO 13485 standards. The Barkey plasmatherm is also FDA 510(k) cleared.
Here you can learn more about the Barkey plasmatherm.
 Querschnitts-Leitlinien zur Therapie mit Blutkomponenten und Plasmaderivaten,4. überarbeitete und aktualisierte Auflage 2014, Herausgegeben vom Vorstand der Bundesärztekammer auf Empfehlung des wissenschaftlichen Beirats
 Lockwood, W.B., Leonard, J., Liles, S.L. (2008): Storage, Monitoring, Pretransfusion Processing, and Distribution of Blood components, in AABB Technical Manual Chapter 9, S. 292
 O’Shaughnessy, D.F., Atterbury, C., Bolton Maggs, P., Murphy, M., Thomas, D., Yates, S., Williamson, L.M. (2004):Guidelines for the use of fresh-frozen plasma, cryoprecipitate and cryosupernatant, The British Society for Heamatology, 126, 11-28, S. 18
 Dumont, L.J., Papari, M., Aronson, C.A., Dumont, D.F. (2014): Whole-blood collection and component processing, in: Fung, M.K., Grossman, B.J., Hillyer, C.D., Westhoff, C.M., AABB Technical Manual. 18th edition, Bethesda, S. 151
Minimal risk of infection is ensured by the Barkey dry tempering process. Nosocomial germs, as they can occur for example in open water baths, represent a potential danger for the patient or user, and can unnecessarily burden the hospital budget,. The spokesman of the Deutsche Gesellschaft für Krankenhaushygiene, Dr. Klaus Zastrow, has determined in a study that the profitability of a clinic can be significantly improved by reducing nosocomial infections. In Germany alone, there are 800,000-1,000,000 hospital infections annually, of which approx. 30-50% are preventable. The length of stay in hospital can be up to seven days longer, with a daily nursing charge of approx. 300€. Zastrow calculates that a reduction of 40% of infections could save approximately 400,000€ per year per clinic. The Barkey plasmatherm thus makes a valuable contribution to eliminating infections. The British Society of Heamatology recommends the use of a dry tempering control device to thaw or warm blood and blood products to avoid the risk of infection.
 Pinki, S.S., Mohan, G., Rafi, A.M., Innah, S.J., Thomas, T. (2017): Rapid dry plasma thawing system: An alternative to conventional thawing baths, in: Asian Journal of Transfusion Science, Vol. 11, No. 2, S. 150
 Walther-Wenke, G., Däubener, W., Heiden, M., Hoch, J., Hornei, B., Volkers, P. Wirsing von König, C.H., Working party on bacteria safety in transfusion medicine of the National Advisory Committee Blood of the Federal Ministry of Health (Arbeitskreis Blut) (2011): Effect of safety measures on bacterial contamination rates of blood components in Germany, in: Transfusion Medicine and Hemotherapy, Vol. 38, No. 4, S. 234
 Taylor, R.W., Manganaro, L., O’Brien, J., Trottier, S.J., Parkar, N., Veremakis, C. (2002): Impact of allogenic packed red blood cell transfusion on nosocomial infection rates in the critically ill patient, in: Critical Care Medicine, Vol. 30, No. 10, S. 2251 f.
 Deutsche Gesellschaft für Krankenhaushygiene e.V. (DGKH) (2018): „Und zack – ist die Nadel auch schon drin!“, Interview mit Prof. Klaus-Dieter Zastrow, https://www.krankenhaushygiene.de/informationen/presseinformationen/pressearchiv/101, 31.01.2018
 Pinki, S.S., et al. (2017): Rapid dry plasma thawing system: An alternative to conventional thawing baths, in: Asian Journal of Transfusion Science, Vol. 11, No. 2, S. 150
 O’Shaughnessy, D.F, et. al. (2004): Guidelines for the use of fresh-frozen plasma, cryoprecipitate and cryosupernatant, The British Society for Heamatology, 126, 11-28, S. 18
The Barkey plasmatherm is designed for simple and intuitive handling and offers additional operating features. Menu navigation in local language, short-cut programs and optional documentation software are available for selection.
The short-cut programs for Plasma and User can be customized in terms of time and temperature (in compliance with the current FDA and AABB guidelines). The Barkey plasmatherm can be used in a temperature range between +30C and +37C. The Blood prorgamm is fixed to +37C.
The cleaning effort for the user is also minimal. A water change of the Barkey plasmatherm is only necessary once a year. Regular wipe disinfection of the temperature chamber is recommended.
Optionally, the Barkey plasmatherm offers the possibility of electronic documentation of the warming or thawing process. The Barkey TCP Logging Tool can be used to document barcodes, device users, time and temperature of the executed program, etc.
Barkey has established a globally active partner network. Our distributors are regularly informed and trained in technical trainings about the current status of a Technical Safety Inspection (TSI). Our international partners and the Barkey Service Team are happy to answer your questions about service and maintenance.
The Barkey plasmatherm has a two-year manufacturer’s warranty.
Here you will find the contact of our Barkey US Service Center in Woburn, MA.
We look forward to hearing from you!
|Power supply||100-120 V 50/60 Hz|
|Set temperature||+30C to max. +37C|
|Dimensions||13.5 x 23.6 x 12.6 in (WxDxH)|
|Weight without filling||40 lbs|
|Filling quantity||9 l|
|Certification||ISO 13485, FDA, UL|
- Documentation software Barkey TCP Logging Tool (for Ethernet-port only)
- Barcode scanner
- Log printer